Welch Allyn Home 1700 (H-BP100SBP)
The technology used in the Welch Allyn Home 1700 (H-BP100SBP), to measure blood pressure, has passed in a clinical validation study, in a specific population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Recommended in certain condition(s)||Validation in specific condition(s). See publications for details.|
|Stride BP||Self-measurement for adults (Preferred)||Published evidence|
|US Blood Pressure Validated Device Listing||Self-measurement||Published evidence|
|Amazon Best Sellers (US)||2019||A (3.8 / 5)|
Alpert BS. Validation of the Welch Allyn Home blood pressure monitor with professional SureBP algorithm with a special feature of accuracy during involuntary (tremor) patient movement. Blood Press Monit. 2019 Apr;24(2):89-92. Epub: 2019 Feb 4. doi: 10.1097/MBP.0000000000000365. PMID: 30724792.
81060-2:2013 - Pass 85 adult subjects whose oscillometric pressures were augmented with simulator-produced tremors.