Hartmann Tensoval Duo Control II
The technology used in the Hartmann Tensoval Duo Control II, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
|ESH (Europe)||Self-measurement||Published evidence|
|Stride BP||Self-measurement for adults (Preferred)||Published evidence|
de Greeff A, Shennan AH. Validation of the Tensoval Duo Control II blood pressure monitor for clinic use and self-measurement according to the British Hypertension Society protocol and the European Society of Hypertension International Protocol Revision 2010. Blood Press Monit. 2013 Jun;18(3):161-6. doi: 10.1097/MBP.0b013e328360fb52. PMID: 23640067.
ESH-IP:2010 - Pass General population