Pic Solution One Check
The technology used in the Pic Solution One Check, to measure blood pressure, has been compared to a clinically validated device but not according to a protocol compliant with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|ESH (Europe)||Self-measurement||Published evidence|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Pic Solution My Check
Germano G, Psimenos A, Sarullo F, Venditti A, Pecchioli V, Asmar R. Validation of four automatic devices for self-measurement of blood pressure according to the International Protocol: the Pic Indolor Personal Check, Comfort Check, My Check and Travel Check. Blood Press. 2009 Oct;18(1):15-23. doi: 10.3109/08037050903262496. PMID: 20050101.
ESH-IP:2002 - Pass General population