Omron MX3 Plus (HEM-742-E)
The technology used in the Omron MX3 Plus (HEM-742-E), to measure blood pressure, has passed in clinical validation studies, in general populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||★ Recommendation||Older clinical validation; older protocol; multiple clinical validations|
|BIHS (UK and IRL)||Previous recommendation, now archived||Published evidence|
|ESH (Europe)||Self-measurement for adults and children||Published evidence|
|Stride BP||Self-measurement for adults||Published evidence|
Coleman A, Freeman P, Steel S, Shennan A. Validation of the Omron MX3 Plus oscillometric blood pressure monitoring device according to the European Society of Hypertension international protocol. Blood Press Monit. 2005 Jun;10(3):165-8. PMID: 15923819.
ESH-IP:2002 - Pass General population
Christofaro DG, Casonatto J, Polito MD, Cardoso JR, Fernandes R, Guariglia DA, Gerage AM, de Oliveira AR. Evaluation of the Omron MX3 Plus monitor for blood pressure measurement in adolescents. Eur J Pediatr. 2009 Nov;168(11):1349-54. Epub: 2009 Feb 17. doi: 10.1007/s00431-009-0936-x. PMID: 19221789.
BHS:1993 - Pass (A/A) General population
SP10:1996 - Pass General population
Ware A, Stevens R, Selwood M, Fleming S. Accuracy of blood pressure monitors available in high street pharmacies. Blood Press Monit. 2016 Feb;21(1):59-61. doi: 10.1097/MBP.0000000000000151. PMID: 26427055.
Measurements from the Omron MX3 Plus (HEM-742-E) are compared to those from the Omron BPM tester PA-350, though not in a formal validation.