Omron MX2 Basic (HEM-742-E2)
The technology used in the Omron MX2 Basic (HEM-742-E2), to measure blood pressure, has been compared to a clinically validated device but not according to a protocol compliant with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|BIHS (UK and IRL)||Previous recommendation, now archived||Manufacturer declaration of equivalence|
|ESH (Europe)||Self-measurement||Published evidence|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Anwar YA, Giacco S, McCabe EJ, Tendler BE, White WB. Evaluation of the efficacy of the Omron HEM-737 IntelliSense device for use on adults according to the recommendations of the Association for the Advancement of Medical Instrumentation. Blood Press Monit. 1998 Aug;3(4):261-265. PMID: 10212364.
SP10:1992 - Pass General population
Omron M2 Compact (HEM-7102-E)
Asmar R, Khabouth J, Topouchian J, El Feghali R, Mattar J. Validation of three automatic devices for self-measurement of blood pressure according to the International Protocol: The Omron M3 Intellisense (HEM-7051-E), the Omron M2 Compact (HEM 7102-E), and the Omron R3-I Plus (HEM 6022-E). Blood Press Monit. 2010 Feb;15(1):49-54. doi: 10.1097/MBP.0b013e3283354b11. PMID: 20032779.
ESH-IP:2002 - Pass General population