Omron M1 Compact (HEM-4022-E)
The technology used in the Omron M1 Compact (HEM-4022-E), to measure blood pressure, has been compared to a clinically validated device but not according to a protocol compliant with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|BIHS (UK and IRL)||Previous recommendation, now archived||Manufacturer declaration of equivalence|
|ESH (Europe)||Self-measurement||Published evidence|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Omron HEM Solar (HEM-4500-Sole)
Parati G, Kilama MO, Faini A, Facelli E, Ochen K, Opira C, Mendis S, Wang J, Atkins N, O'Brien E. A new solar-powered blood pressure measuring device for low-resource settings. Hypertension. 2010 Dec;56(6):1047-53. Epub: 2010 Nov 8. doi: 10.1161/HYPERTENSIONAHA.110.160408. PMID: 21059998. Available from: hyper.ahajournals.org.
ESH-IP:2002 - Fail General population (Note: Passed SBP)
Study includes field test in LR settings
Omron M1 Plus (HEM-4011C-E)
Belghazi J, El Feghali RN, Moussalem T, Rejdych M, Asmar RG. Validation of four automatic devices for self-measurement of blood pressure according to the International Protocol of the European Society of Hypertension. Vasc Health Risk Manag. 2007;3(4):389-400. PMID: 17969368. Available from: PMC2291343.
ESH-IP:2002 - Pass General population