Omron BP760N (HEM-7320-Z)
The accuracy of the technology used in the Omron BP760N (HEM-7320-Z), to measure blood pressure, is certified by Medaval Ltd.
|Medaval||★★★★ Certified Accuracy||Recent clinical validation; recent protocol with certified adherence|
|ESH (Europe)||Self-measurement||Published evidence|
|Stride BP||Self-measurement for adults and pregnancy (Preferred)||Published evidence|
|Amazon Best Sellers (US)||2019||A (4.2 / 5)|
Takahashi H, Yoshika M, Yokoi T. Validation of two automatic devices for the self-measurement of blood pressure according to the ANSI/AAMI/ISO81060-2:2009 guidelines: the Omron BP765 (HEM-7311-ZSA) and the Omron BP760N (HEM-7320-Z). Vasc Health Risk Manag. 2015 Jan;9(11):49-53. doi: 10.2147/VHRM.S72438. PMID: 25657587.
81060-2:2009 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Omron BP-760N (HEM-7320-Z) upper arm monitor, as validated according to the AAMI/ANSI/ISO 81060-2:2013 standard for a general study in adults. Medical Device Assessment. 2016 Aug 5;2016(1624) 4 p. Epub: 2019 Jan 31. Available from:
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the Omron BP760N (HEM-7320-Z), is certified by Medaval Ltd.