The technology used in the InBody BPBIO750, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BIHS (UK and IRL)||Public use||Manufacturer declaration of equivalence|
Ntineri A, Prapa S, Bountzona I, Menti A, Stergiou GS. Validation of the single-cuff oscillometric blood pressure monitor InBody BPBIO750 for public spaces according to the Association for the Advancement of Medical Instrumentation/European Society of Hypertension/International Organization for Standardization Universal Standard. Blood Press Monit. Epub: 2020 Dec 14. doi: 10.1097/MBP.0000000000000507. PMID: 33323723.
81060-2:2018 - Pass General population
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Kollias A, Stambolliu E, Kyriakoulis KG, Papadatos SS, Stergiou GS. Validation of the single-cuff oscillometric blood pressure monitor InBody BPBIO320 for public use according to the 2010 European Society of Hypertension International Protocol. Blood Press Monit. 2019 Feb;24(1):30-32. Epub: 2018 Dec 7. doi: 10.1097/MBP.0000000000000359. PMID: 30531495.
ESH-IP:2010 - Pass General population