Grandway G.LAB MD2231
The technology used in the Grandway G.LAB MD2231, to measure blood pressure, is among the best rated for accuracy by Medaval Ltd. It has passed in clinical validation studies, between general and specific populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||★★★ Recommendation||Recent clinical validation; recent protocol; multiple clinical validations|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
|Stride BP||Self-measurement||Published evidence|
|AMA VDL Criteria||Satisfies stated criteria||510(k) K133618 plus 81060-2:2013 and BHS:1993 validations|
|MDR Criteria||Self-measurement||Published evidence|
Jiao Y, Guan Q, Wu L, Wang C, Cao L. Validation of the G.LAB MD2231 digital automatic wrist blood pressure monitor according to multiple protocols. Blood Press Monit. 2017 Aug;22(4):226-9. Epub: 2017 Mar 21. doi: 10.1097/MBP.0000000000000256. PMID: 28328581.
ESH-IP:2010 - Pass General population
81060-2:2013 - Pass General population
BHS:1993 - Pass (A/A) General population