The technology used in the Dixtal DX-2710, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Legacy approval||Older clinical validation; older protocol|
Mano GM, Souza VF, Pierin AM, Lima JC, Ignes EC, Ortega KC, Mion D Jr. Avaliação do Aparelho Automático Oscilométrico de Medida da Pressão Arterial DIXTAL DX-2710 pelos Protocolos de Validação da British Hypertension Society (BHS) e Association for the Advancement of Medical Instrumentation (AAMI). [Assessment of the DIXTAL DX-2710 automated oscillometric device for blood pressure measurement with the validation protocols of the British Hypertension Society (BHS) and the Association for the Advancement of Medical Instrumentation (AAMI).] Arq Bras Cardiol. 2002 Dec;79(6):606-10, 601-5. PMID: 12532244. Available from: publicacoes.cardiol.br. In English, Portuguese.
BHS:1993 - Pass (A/B) General population
SP10:1992 - Pass General population