The technology used in the Dinamap 8100, to measure blood pressure, has passed in a clinical validation study, in a specific population, according to a recognised standard protocol, but has failed in a validation study, in a general population, also according to a recognised standard protocol, as published in a peer-reviewed publications. It has also been assessed, in a general population, according to a non-standard protocol.
|Medaval||Recommended in certain condition(s)||Validation in specific condition(s). See publications for details.|
Ornstein S, Markert G, Litchfield L, Zemp L. Evaluation of the DINAMAP blood pressure monitor in an ambulatory primary care setting. J Fam Pract. 1988 May;26(5):517-21. PMID: 3367116.
Ad Hoc protocol General population (Note: Compared to Hawksley RZ, fail reported; n=80)
O'Brien E, Mee F, Atkins N, O'Malley K. Short report: accuracy of the Dinamap portable monitor, model 8100 determined by the British Hypertension Society protocol. J Hypertens. 1993 Jul;11(7):761-3. PMID: 8228196.
BHS:1993 - Fail (B/D) General population
SP10:2002 - Fail General population (Note: Failed DBP)
O'Brien E. Accuracy of the Dinamap 8100. Am J Hypertens. 1997 Jan;10(1):143-4. PMID: 9008262.
BHS:1993 - Fail General population
Barker ME, Shiell AW, Law CM. Evaluation of the Dinamap 8100 and Omron M1 blood pressure monitors for use in children. Paediatr Perinat Epidemiol. 2000 Apr;14(2):179-86. PMID: 10791663.
BHS:1993 - Fail Children (n=55) (Note: No grades in Abstract)
Jin RZ, Donaghue KC, Fairchild J, Chan A, Silink M. Comparison of Dinamap 8100 with sphygmomanometer blood pressure measurement in a prepubertal diabetes cohort. J Paediatr Child Health. 2001 Dec;37(6):545-9. PMID: 11903832.
BHS:1993 - Pass (B/B) Children with Diabetes Mellitus
SP10:1996 - Pass Children with Diabetes Mellitus
O'Brien E, Atkins N. Inaccuracy of the Dinamap 8100 portable monitor. Lancet. 1997 Apr 5;349(9057):1026. PMID: 9100649.
The publication describes the use of the Dinamap 8100 in a scientific study.
O'Brien E, Atkins N. Accuracy of the Dinamap Portable Monitor, model 8100: a review of the evidence for accuracy. Blood Press Monit. 1997 Dec;2(1):31-33. PMID: 10234088.
The publication is a letter or correspondence in which the Dinamap 8100 is referenced.
Amoore JN, Geake WB. Evaluation of the Critikon 8100 and Spacelabs 90207 non-invasive blood pressure monitors using a test simulator. J Hum Hypertens. 1997 Mar;11(3):163-9. PMID: 9175568.
Measurements from the Dinamap 8100 are compared to those from a simulator, though not in a formal validation.
Amoore JN. A comparative evaluation of the DINAMAP 8100 and DINAMAP Compact TS using a non-invasive blood pressure simulator. Blood Press Monit. 1998 Oct;3(5):309-314. PMID: 10212371.
Measurements from the Dinamap 8100 are compared to those from a simulator and the Dinamap Compact TS, though not in a formal validation.
Papadopoulos G, Mieke S, Elisaf M. Assessment of the performances of three oscillometric blood pressure monitors for neonates using a simulator. Blood Press Monit. 1999 Feb;4(1):27-33. PMID: 10362888.
Neonate, simulator v SL M-90426 & HP M 1008B