Arkray GlucoCard X-Meter
The technology used in the Arkray GlucoCard X-Meter, to measure blood glucose, has passed in clinical validation studies, in general populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||★★ Recommendation||Recent clinical validation; older protocol; multiple clinical validations|
Weitgasser R, Hofmann M, Gappmayer B, Garstenauer C. New, small, fast acting blood glucose meters--an analytical laboratory evaluation. Swiss Med Wkly. 2007 Sep 22;137(37-38):536-40. PMID: 17990144.
ADA:1996 - Pass General population
Freckmann G, Baumstark A, Jendrike N, Zschornack E, Kocher S, Tshiananga J, Heister F, Haug C. System accuracy evaluation of 27 blood glucose monitoring systems according to DIN EN ISO 15197. Diabetes Technol Ther. 2010 Mar;12(3):221-31. doi: 10.1089/dia.2009.0128. PMID: 20151773.
15197:2003 - Pass General population
SKUP. Report from the evaluation SKUP/2006/50: Glucocard X Meter with Glucocard X-Sensor test strips (Arkray Global Business), a system for blood glucose measurement. Bergen, Norway: SKUP; 2006. 51 p. Available from: skup.org.
15197:2003 - Pass General population (Note: BLS-recorded measurements)
15197:2003 - Pass General population (Note: Subject-recorded measurements; less accurate than BLS recorded measurements)
Arkray Glucocard X-Sensor test strips
권민정, 이수연 [Kwon MJ, Lee SY.] 혈당측정기 GLUCOCARD X-METER 평가 [Evaluation of GLUCOCARD X-METER glucose monitoring system.] 대한진단검사의학회지 [Korean J Lab Med]. 2008 Feb;28(1):8-15. doi: 10.3343/kjlm.2008.28.1.8. PMID: 18309250. Available from: synapse.koreamed.org. In Korean.
Blind comparisons to the Arkray Super Glucocard II, the Roche Accu-Check Active (Probably V1) and the JnJ Lifescan Onetouch Ultra. It is unclear from the publication which Accu-Chek Active device was used in the study. The date suggests that the version Medaval has indicated as "V1" was used.