The technology used in the A&D UA-1200BLE, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|Stride BP||Self-measurement for adults (Preferred)||Published evidence|
|US Blood Pressure Validated Device Listing||Self-measurement||Published evidence plus 510(k) K163683|
|MDR Criteria||Self-measurement||Published evidence|
Kario K, Saito K, Sato K, Hamasaki H, Suwa H, Okura A, Hoshide S. Validation of the A&D BP UA-1200BLE device for home blood pressure measurement according to the ISO 81060-2: 2013 standard. Blood Press Monit. 2018 Dec;23(6):312-314. Epub: 2018 Sep 7. doi: 10.1097/MBP.0000000000000347. PMID: 30199378.
81060-2:2013 - Pass General population