The technology used in the A&D TM-2656, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★ Recommendation||Recent clinical validation; older protocol|
|BIHS (UK and IRL)||Public use||Published evidence|
|ESH (Europe)||Professional use||The BIHS recommendation|
|Japanese Society of Hypertension||Self-measurement and professional use (2016-2019)||Published evidence|
|Stride BP||Public use for adults (Preferred)||Published evidence|
Zeng WF, Liu M, Kang YY, Li Y, Wang JG. Validation of the fully automated A&D TM-2656 blood pressure monitor according to the British Hypertension Society Protocol. Blood Press Monit. 2013 Aug;18(4):223-6. doi: 10.1097/MBP.0b013e3283624abd. PMID: 23681206.
BHS:1993 - Pass (A/A) General population