iHealth CardioMed (ABP100)
OBL: iHealthlabs Europe SAS, 36 Rue de Ponthieu, 75008 Paris, FRANCE.
OEM: Andon Health Co. Ltd., 3 Jinping Road, Ya'an Street, Nankai District, Tianjin 300190, CHINA.
Primary Client Use:
Intended for ambulatory blood pressure monitoring
Intermittent measurements at specified intervals or times
Availability according to Countries or Regions:
Ireland and South Africa
The iHealth CardioMed (ABP100) is an ambulatory blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for ambulatory blood pressure monitoring.
The technology used in the iHealth CardioMed (ABP100), to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|BP||Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BP||Stride BP||ABPM for adults (Preferred)||Published evidence|
|BP||MDR Criteria||ABPM||Published evidence|
Kollias A, Ntineri A, Kyriakoulis KG, Stambolliu E, Stergiou GS. Validation of the iHealth ambulatory blood pressure monitor in adults according to the American National Standards Institute/Association for the Advancement of Medical Instrumentation/International Organization for Standardization standard. Blood Press Monit. 2018 Apr;23(2):115-6. Epub: 2018 Feb 05. doi: 10.1097/MBP.0000000000000314. PMID: 29406327.
81060-2:2013 - Pass General population