Guangdong Transtek Medical Electronics Co. Ltd., Zone A, 105 Dongli Road, Torch Development District, Zhongshan 528437, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The Transtek LS808-B is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use.
The technology used in the Transtek LS808-B, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|BP||Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BP||ESH (Europe)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement for adults (Preferred)||Published evidence|
Liu ZH, Liu XY, Wu WJ. Validation of Transtek LS808-B for self/home measurement according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2016 Dec;21(6):352-5. doi: 10.1097/MBP.0000000000000212. PMID: 27622888.
ESH-IP:2010 - Pass General population