RisingSun, Beijing, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
Not made available
The RisingSun RS-651 is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use.
The technology used in the RisingSun RS-651, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|BP||Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BP||ESH (Europe)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement and professional use for adults (Preferred)||Published evidence|
She J, Guan X, Liu Y, Xiang H Validation of the RisingSun RS-651 Blood Pressure Monitor Based on Auscultation in Adults According to the ANSI/AAMI/ISO 81060-2:2013 Standard. J Clin Hypertens (Greenwich). 2016 Dec;18(12):1279-83. Epub: 2016 Jun 7. doi: 10.1111/jch.12859. PMID: 27271472. Available from: onlinelibrary.wiley.com.
ESH-IP:2010 - Pass General population