Shenzhen Pango Electronic Co. Ltd., 1st Industrial Zone, 25 Fenghuang Lu, Xikeng, Henggang, Longgang District, Shenzhen, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The PangaO PG-800B26 is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 3-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use.
The technology used in the PangaO PG-800B26, to measure blood pressure, is among the best rated for accuracy by Medaval Ltd. It has passed in two clinical validation studies, between general and specific populations, according to recognised standard protocols, as published in peer-reviewed publications.
|BP||Medaval||★★★ Recommendation||Recent clinical validation; recent protocol; multiple clinical validations|
|BP||BIHS (UK and IRL)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement for adults (Preferred)||Published evidence|
|BP||MDR Criteria||Self-measurement||Published evidence|
Chen L, Li J, Wen J, Guo C, Zhang J, Yu Z. Validation of the Pangao PG-800B26 upper arm blood pressure monitor in the general population according to the European Society of Hypertension and the British Hypertension Society protocols. Blood Press Monit. 2018 Feb;23(1):41-4. Epub: 2017 Sep 11. doi: 10.1097/MBP.0000000000000289. PMID: 28902676.
ESH-IP:2010 - Pass General population
BHS:1993 - Pass (A/A) General population
81060-2:2013 - Pass General population