PangaO PG-800A36 (V1)
Shenzhen Pango Electronic Co. Ltd., 1st Industrial Zone, 25 Fenghuang Lu, Xikeng, Henggang, Longgang District, Shenzhen, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The PangaO PG-800A36 (V1) is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 3-star Medaval rating. Blood pressure measurements are taken from the wrist. It is intended for self-measurement and home use.
The technology used in the PangaO PG-800A36 (V1), to measure blood pressure, is among the best rated for accuracy by Medaval Ltd. It has passed in two clinical validation studies, between general and specific populations, according to recognised standard protocols, as published in peer-reviewed publications.
|BP||Medaval||★★★ Recommendation||Recent clinical validation; recent protocol; multiple clinical validations|
|BP||BIHS (UK and IRL)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement for adults||Published evidence|
Zhao H, Qiao W, Zhang R, Cui P, Hou F, Zhang W. Validation of the Pangao PG-800A36 automatic wrist blood pressure monitor according to the European Society of Hypertension and the British Hypertension Society protocols. Blood Press Monit. 2018 Feb;23(1):37-40. Epub: 2017 Sep 18. doi: 10.1097/MBP.0000000000000288. PMID: 28926362 .
ESH-IP:2010 - Pass General population
BHS:1993 - Pass (A/A) General population