Nihon-Kohden Vismo PVM-2701
Distributer: Nihon Kohden UK Ltd., Unit 118 Trident Court, 1 Oakcroft Road Chessington, Surrey KT9 1BD, UNITED KINGDOM.
Other: Nihon Kohden Corporation, 1-31-4 Nishiochiai, Shinjuku-ku, Tokyo 161-8560, JAPAN.
Blood pressure, Arrhythmias, SpO2, Temperature, ECG, Respiration
Primary Client Use:
Intended for patient monitoring
BP: Upper Arm; SpO2: Finger
BP & SpO2: Continuous measurement
The Nihon-Kohden Vismo PVM-2701 is a patient monitor. Its blood pressure measurement technology has been proven to be accurate, with a 1-star Medaval rating and information regarding the accuracy of its oxygen saturation measurement technology does not appear to be available. Blood pressure measurements are taken from the upper arm and oxygen saturation measurements are taken from the finger. It is intended for bedside patient monitoring.
The technology used in the Nihon-Kohden Vismo PVM-2701, to measure blood pressure, has passed in a clinical validation study, in a general population including children, according to a recognised standard protocol, as published in a peer-reviewed publication. There appears to be no peer-reviewed clinical validation information available on the technology used in the Nihon-Kohden Vismo PVM-2701 to measure peripheral oxygen saturation.
||Recent clinical validation; older protocol
||Professional use for children
||Hospital use for adults and children
Nihon-Kohden Vismo PVM-2701J
, Nihon-Kohden BSM-1700J
, Nihon-Kohden BSM-3000J
, Nihon-Kohden BSM-6000J
, Nihon-Kohden CSM-1500J
, Nihon-Kohden CSM-1700J
, Nihon-Kohden CSM-1900J
, Nihon-Kohden PVM-2700J
, Nihon-Kohden TEC-8000J
, Nihon-Kohden ZM-540PGJ
, Nihon-Kohden ZM-940PGJ
, Nihon-Kohden GZ-140PJ
, Nihon-Kohden PVM-9900J
, Nihon-Kohden RMC-5000J
, Nihon-Kohden SVM-7000J
, Nihon-Kohden TEC-5000J
, Nihon-Kohden ZS-640J
, Nihon-Kohden ZS-940PGJ
, Nihon-Kohden Vismo PVM-4000J
, Nihon-Kohden Cardiolife EMS EMS-1052J
Legend: J JHS Implied Equivalence
Alpert BS. Validation of the Nihon Kohden PVM-2701/Impulse-1 automated device by both AAMI (2002) and ISO standards testing. Blood Press Monit. 2012 Oct;17(5):207-9. doi: 10.1097/MBP.0b013e328359c48e. PMID: 22968109.
81060-2:2009 - Pass Adults (n=69) and children (n=41)
SP10:2002 - Pass Adults (n=69) and children (n=41)