Shenzhen Pump Medical System Co. Ltd., 2/F - No 7 Sinosteel Building, Maqueling Estate, Hi-Tech Industrial Park, Nanshan, 518057 Shenzhen, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The Bpump BF1112 is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use.
The technology used in the Bpump BF1112, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|BP||Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BP||BIHS (UK and IRL)||Self-measurement||Published evidence|
|BP||ESH (Europe)||Self-measurement||Published evidence|
|BP||Stride BP||Self-measurement for adults (Preferred)||Published evidence|
Chen Q, Kang YY, Li Y, Wang JG. Validation of the BPUMP BF1112 upper-arm blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2017 Apr;22(2):105-8. Epub: 2017 Jan 7. doi: 10.1097/MBP.0000000000000236. PMID: 28072599.
ESH-IP:2010 - Pass General population