Beneware 24h-ABPM (ABP-021)
Hangzhou Beneware Medical Equipment Co. Ltd., Room 1001, 2 Hydra Blk, Singapore Hangzhou Science & Technology Park, Xia Sha Disctrict, 310018 Hangzhou, CHINA.
Primary Client Use:
Intended for ambulatory blood pressure monitoring
Intermittent measurements at specified intervals or times
The Beneware 24h-ABPM (ABP-021) is an ambulatory blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for ambulatory blood pressure monitoring.
The technology used in the Beneware 24h-ABPM (ABP-021), to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|BP||Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BP||BIHS (UK and IRL)||NOT recommended||Published evidence|
|BP||Stride BP||ABPM for adults (Preferred)||Published evidence|
Pereira T, Guimarães J. Validation of the Beneware model ABP-021 ambulatory blood pressure monitor according to the revised 2010 European Society of hypertension international protocol. Blood Press Monit. 2018 Aug;23(4):210-3. Epub: 2018 Jun 11. doi: 10.1097/MBP.0000000000000330. PMID: 29894314.
ESH-IP:2010 - Pass General population