BIOS BD240 - 2019 (Precision 8.0 - Protocol 7D)
Precision 8.0 - Protocol 7D
Distributer: Thermor Ltd., 16975 Leslie Street, Newmarket, Ontario L3Y 9A1, CANADA.
OBL: BIOS Medical, 16975 Leslie Street, Newmarket, Ontario L3Y 9A1, CANADA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
Availability according to Countries or Regions:
The BIOS BD240 - 2019 (Precision 8.0 - Protocol 7D) is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm. It is intended for self-measurement and home use.
The technology used in the BIOS BD240 - 2019 (Precision 8.0 - Protocol 7D), to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
||Recent clinical validation; recent protocol
||Self-measurement for adults (Preferred)
Ringrose JS, Wood PW, Diaz-Carvallo F, Wirzba M, Padwal R. Validation of the BIOS BD240 home blood pressure monitor in a multi-racial, multi-morbid Canadian population according to the International Standards Organization 2018 Protocol. Blood Press Monit. 2020 Jun;25(3):167-170. Epub: 2020 Mar 16. doi: 10.1097/MBP.0000000000000444. PMID: 32187037.
81060-2:2019 - Pass General population