While the technology used in the Unimedic Tensioday, to measure blood pressure, has been declared as being equivalent to that used in another device, not only has no evidence has been published to show that the devices were compared according to a protocol compliant with (EU) 2017/745 and MEDDEV 2.7/1 rev 4 but the results of the assessment on which the declaration has been based have not been published either and cannot be verified. However, there are validation studies proving clinical validation of the same measurement technology, according to other equivalence claims.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|BIHS (UK and IRL)||ABPM||Implied equivalence to the Tensiomed Tensioday|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Németh Z, Móczár K, Deák G. Evaluation of the Tensioday ambulatory blood pressure monitor according to the protocols of the British Hypertension Society and the Association for the Advancement of Medical Instrumentation. Blood Press Monit. 2002 Jun;7(3):191-7. PMID: 12131077.
BHS:1993 - Pass (A/A) General population
SP10:1992 - Pass General population