The technology used in the Suntech 247, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Legacy approval||Older clinical validation; older protocol|
|BIHS (UK and IRL)||Professional use||Published evidence|
|ESH (Europe)||Professional use||Published evidence|
|Stride BP||Professional use (Preferred)||Published evidence|
Polo Friz H, Facchetti R, Primitz L, Beltrame L, Galbiati V, Ricioppo A, Bombelli M, Sega R. Simultaneous validation of the SunTech 247 diagnostic station blood pressure measurement device according to the British Hypertension Society protocol, the International Protocol and the Association for the Advancement of Medical Instrumentation standards. Blood Press Monit. 2009 Oct;14(5):222-7. doi: 10.1097/MBP.0b013e328330c873. PMID: 19741510.
ESH-IP:2002 - Pass General population
BHS:1993 - Pass (A/A) General population
SP10:2002 - Pass General population