Somno Somnotouch NIBP (TON100)
The technology used in the Somno Somnotouch NIBP (TON100), to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
Bilo G, Zorzi C, Ochoa Munera JE, Torlasco C, Giuli V, Parati G. Validation of the Somnotouch-NIBP noninvasive continuous blood pressure monitor according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2015 Oct;20(5):291-4. Epub: 2015 Apr 28. doi: 10.1097/MBP.0000000000000124. PMID: 25932885.
ESH-IP:2010 - Pass General population
Manning LS, Robinson TG, Panerai RB. The SOMNOtouch device as a novel method for measuring short-term blood pressure variability: a comparison with the Finometer. Blood Press Monit. 2015 Dec;20(6):361-8. doi: 10.1097/MBP.0000000000000128. PMID: 26154853.
Measurements from the Somno Somnotouch NIBP (TON100) are compared to those from the Finapres Finometer Midi, though not in a formal validation.
Golberg M, Ruiz-Rivas J, Polani S, Beiderman Y, Zalevsky Z. Large-scale clinical validation of noncontact and continuous extraction of blood pressure via multipoint defocused photonic imaging. Appl Opt. 2018 Mar 1;57(7):B45-B51. doi: 10.1364/AO.57.000B45. PMID: 29522038.
The publication describes the Somno Somnotouch NIBP (TON100) being used as a reference device for comparison with .