Rossmax CF175f (V1)
The accuracy of the technology used in the Rossmax CF175f (V1), to measure blood pressure, is certified by Medaval Ltd.
|Medaval||★★★★ Certified Accuracy||Recent clinical validation; recent protocol with certified adherence|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
|ESH (Europe)||Self-measurement||Published evidence|
|Stride BP||Self-measurement for adults (Preferred)||Published evidence|
Zhang L, Kang YY, Zeng WF, Li Y, Wang JG. Validation of the Rossmax CF175 upper-arm blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2015 Apr;20(2):104-7. doi: 10.1097/MBP.0000000000000089. PMID: 25350783.
ESH-IP:2010 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Rossmax CF175 upper arm monitor, as validated according to the European Society of Hypertension International Protocol revision 2010. Medical Device Assessment. 2016 Aug 5;2016(1621) 4 p. Epub: 2019 Jan 31. Available from:
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the Rossmax CF-175F, is certified by Medaval Ltd.