Pic Solution My Check
The technology used in the Pic Solution My Check, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Legacy approval||Older clinical validation; older protocol|
|ESH (Europe)||Self-measurement||Published evidence|
|Stride BP||Self-measurement for adults||Published evidence|
Germano G, Psimenos A, Sarullo F, Venditti A, Pecchioli V, Asmar R. Validation of four automatic devices for self-measurement of blood pressure according to the International Protocol: the Pic Indolor Personal Check, Comfort Check, My Check and Travel Check. Blood Press. 2009 Oct;18(1):15-23. doi: 10.3109/08037050903262496. PMID: 20050101.
ESH-IP:2002 - Pass General population