The technology used in the Omron HEM-6022, to measure blood pressure, has been compared to a clinically validated device but the results of the equivalence study have not been published and cannot be verified; nor can the protocol be checked for compliance with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|Japanese Society of Hypertension||Self-measurement (2018)||Implied equivalence to the Omron M2 Compact (HEM-7102-E), M3 Intellisense (HEM-7051-E), R3-I Plus (HEM-6022-E)|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Omron R3 (HEM-6021-E)
Eckert S, Gleichmann U, Zagorski O, Klapp A. Validation of the OMRON R3 blood pressure self-measuring device through simultaneous comparative invasive measurements according to protocol 58130 of the German Institute for Validation. Blood Press Monit. 1997 Aug;2(4):189-192. PMID: 10234115.
58130:1995 - Pass General population (Note: (n=75/15))
Dieterle T, Battegay E, Bucheli B, Martina B. Accuracy and 'range of uncertainty' of oscillometric blood pressure monitors around the upper arm and the wrist. Blood Press Monit. 1998;3(6):339-46. PMID: 10212375.
BHS:1993 - Fail General population (Note: No grades in abstract)
SP10:1992 - Fail General population
Watson S, Wenzel RR, di Matteo C, Meier B, Lüscher TF. Accuracy of a new wrist cuff oscillometric blood pressure device: comparisons with intraarterial and mercury manometer measurements. Am J Hypertens. 1998 Dec;11(12):1469-74. PMID: 9880129.
IA protocol - Pass General population (Note: (n=100))
Omron R3-I Plus (HEM-6022-E)
Asmar R, Khabouth J, Topouchian J, El Feghali R, Mattar J. Validation of three automatic devices for self-measurement of blood pressure according to the International Protocol: The Omron M3 Intellisense (HEM-7051-E), the Omron M2 Compact (HEM 7102-E), and the Omron R3-I Plus (HEM 6022-E). Blood Press Monit. 2010 Feb;15(1):49-54. doi: 10.1097/MBP.0b013e3283354b11. PMID: 20032779.
ESH-IP:2002 - Pass General population