Microlife BP 3AS1-2
The technology used in the Microlife BP 3AS1-2, to measure blood pressure, is among the best rated for accuracy by Medaval Ltd. It has passed in three clinical validation studies, between general and specific populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||★★★ Recommendation||Recent clinical validation; recent protocol; multiple clinical validations|
|BIHS (UK and IRL)||Pregnancy||Published evidence|
|MDR Criteria||During pregnancy|
de Greeff A, Nathan H, Stafford N, Liu B, Shennan AH. Development of an accurate oscillometric blood pressure device for low resource settings. Blood Press Monit. 2008 Dec;13(6):342-8. doi: 10.1097/MBP.0b013e32830fd07c. PMID: 19020425.
ESH-IP:2002 - Pass General population
Nathan HL, de Greeff A, Hezelgrave NL, Chappell LC, Shennan AH. An accurate semiautomated oscillometric blood pressure device for use in pregnancy (including pre-eclampsia) in a low-income and middle-income country population: the Microlife 3AS1-2. Blood Press Monit. 2015 Feb;20(1):52-5. doi: 10.1097/MBP.0000000000000086. PMID: 25243711.
BHS:1993 - Pass (B/A) Pregnancy and pre-eclampsia (n=45)
81060-2:2013 - Pass Pregnancy and pre-eclampsia (n=45)
Nathan HL, de Greeff A, Hezelgrave NL, Chappell LC, Shennan AH. Accuracy validation of the Microlife 3AS1-2 blood pressure device in a pregnant population with low blood pressure. Blood Press Monit. 2015 Oct;20(5):299-302. Epub: 2015 May 5. doi: 10.1097/MBP.0000000000000134. PMID: 26020367.
BHS:1993 - Pass (A/A) Low BP pregnancy (n=30)
81060-2:2013 - Pass Low BP pregnancy (n=30)