The technology used in the Lloyds BP11, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Legacy approval||Older clinical validation; older protocol|
|BIHS (UK and IRL)||Previous recommendation, now archived||Published evidence|
|ESH (Europe)||Self-measurement||Published evidence|
|Stride BP||Self-measurement for adults (Preferred)||Published evidence|
Coleman A, Steel S, deGreeff A, Shennan A. Validation of the Lloyds Pharmacy BP11 oscillometric blood pressure monitor according to the international protocol of the European Society of Hypertension. Blood Press Monit. 2010 Jun;15(3):163-6. doi: 10.1097/MBP.0b013e32833713cf. PMID: 20110808.
ESH-IP:2002 - Pass General population