The technology used in the H&L HL888HA, to measure blood pressure, has passed in two clinical validation studies, in general populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||★ Recommendation||Older clinical validation; older protocol; multiple clinical validations|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
|DHL (DE)||General use||Published evidence|
de Greeff A, Shennan AH. The Health and Life device (HL888HA): accuracy assessment in an adult population according to the British Hypertension Society protocol. Blood Press Monit. 2007 Apr;12(2):107-11. PMID: 17353654.
BHS:1993 - Pass (A/A) General population
Tholl U, Lüders S, Bramlage P, Dechend R, Eckert S, Mengden T, Nürnberger J, Sanner B, Anlauf M. The German Hypertension League (Deutsche Hochdruckliga) Quality Seal Protocol for blood pressure-measuring devices: 15-year experience and results from 105 devices for home blood pressure control. Blood Press Monit. 2016 Aug;22(4):197-205. doi: 10.1097/MBP.0000000000000186. PMID: 26998590.
DHL:1999 - Pass General population (Note: 2003)