Dinamap ProCare 400
The technology used in the Dinamap ProCare 400, to measure blood pressure, has passed in two clinical validation studies, in pregnancy populations, according to recognised standard protocols, as published in peer-reviewed publications.
|Medaval||Recommended for pregnancy only||Validation in pregnancy|
|ESH||Professional use during pregnancy||Published evidence|
de Greeff A, Ghosh D, Anthony J, Shennan A. Accuracy assessment of the Dinamap ProCare 400 in pregnancy and preeclampsia. Hypertens Pregnancy. 2010 Jan;29(2):198-205. doi: 10.3109/10641950902968650. PMID: 20367507.
BHS:1993 - Pass (A/A) Pregnancy
BHS:1993 - Pass (A/B) Pre-eclampsia
Brothwell S, Dutton M, Ferro C, Stringer S, Cockwell P. Optimising the accuracy of blood pressure monitoring in chronic kidney disease: the utility of BpTRU. BMC Nephrol. 2013 Oct 10;14:218. doi: 10.1186/1471-2369-14-218. PMID: 24112304. Available from: PMC3852944.
Measurements from the Dinamap ProCare 400 are compared to those from the BPM-100 and the ABPM-04, though not in a formal validation.