The technology used in the Colin HBP-2070, to measure blood pressure, has been compared to a clinically validated device but the results of the equivalence study have not been published and cannot be verified; nor can the protocol be checked for compliance with (EU) 2017/745 and MEDDEV 2.7/1 rev 4.
|Medaval||None||Although equivalence to another device is claimed, this has not been tested to MDR requirements.|
|Japanese Society of Hypertension||Professional use (2016-2018)||Implied equivalence to the Omron M3500 NIBP Module|
Note: This is a provisional list, as equivalence according to EU Regulation 2017/745 (e.g. MEDDEV 2.7/1 rev 4) is not proven. Accordingly, these publications are not used in the assessment of star-ratings.
Omron M3500 NIBP Module
Chahine MN, Assemaani N, Sayed Hassan G, Cham M, Salameh P, Asmar R. Validation of the OMRON M3500 Blood Pressure Measuring Device Using Normal- and High-Speed Modes in Adult and Specific Populations (Obese and Children) According to AAMI Protocol. J Clin Hypertens (Greenwich). 2015 Aug;17(8):622-9. Epub: 2015 Apr 2. doi: 10.1111/jch.12540.. PMID: 25833259. Available from: onlinelibrary.wiley.com.
81060-2:2009 - Pass In Normal-speed mode (Note: 81060-2:2013 criteria fulfilled)
81060-2:2009 - Pass In High-speed mode (Note: 81060-2:2013 criteria fulfilled)