BioCare Diavue Prudential
The technology used in the BioCare Diavue Prudential, to measure blood glucose, has failed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||Not recommended||This device failed clinical validation.|
|Institute of Health Economics (CA)||General use||No evidence provided|
Wolde M, Tarekegn G, Kebede T. Comparative Evaluations of Randomly Selected Four Point-of-Care Glucometer Devices in Addis Ababa, Ethiopia. J Diabetes Sci Technol. 2018 May;12(3):673-9. Epub: 2018 Jan 17. doi: 10.1177/1932296817751747. PMID: 29338325.
15197:2013 - Fail General population
15197:2003 - Fail General population