The technology used in the Avita BPM-64, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
|Medaval||★★ Recommendation||Recent clinical validation; recent protocol|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
|Stride BP||Self-measurement (Preferred)||Published evidence|
Kang YY, Chen Q, Liu CY, Li Y, Wang JG. Validation of the AVITA BPM64 upper-arm blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2018 Feb;23(1):45-8. Epub: 2017 Oct 18. doi: 10.1097/MBP.0000000000000296. PMID: 29049094.
ESH-IP:2010 - Pass General population