The accuracy of the technology used in the Avita BPM63S, to measure blood pressure, is certified by Medaval Ltd.
Kang YY, Zeng WF, Liu M, Li Y, Wang JG. Validation of the AVITA BPM63S upper arm blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2014 Feb;19(1):46-9. doi: 10.1097/MBP.0000000000000014. PMID: 24322871.
ESH-IP:2010 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Avita BPM63S upper arm monitor, as validated according to the European Society of Hypertension International Protocol revision 2010. Medical Device Assessment. Epub: 2016 Aug 5(14):4 p.
The blood pressure measurement technology, used in the Avita BPM-63S, is certified by Medaval Ltd.