The accuracy of the technology used in the A&D UB-543, to measure blood pressure, is certified by Medaval Ltd.
|Medaval||★★★★ Certified Accuracy||Recent clinical validation; recent protocol with certified adherence|
|BIHS (UK and IRL)||Self-measurement||Published evidence|
Fania C, Benetti E, Palatini P. Validation of the A&D BP UB-543 wrist device for home blood pressure measurement according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2015 Aug;20(4):237-40. Epub: 2015 Mar 12. doi: 10.1097/MBP.0000000000000117. PMID: 25768063.
ESH-IP:2010 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the A&D UB-543 wrist monitor, as validated according to the European Society of Hypertension International Protocol revision 2010. Medical Device Assessment. 2016 Aug 5;2016(1626) 4 p. Epub: 2019 Jan 31. Available from:
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the A&D UB-543, is certified by Medaval Ltd.