SunTech Medical Inc., Suite 117, 507 Airport Boulevard, Morrisville, NC 27560-8200, UNITED STATES.
Blood pressure, SpO2, Temperature
Primary Client Use:
Intended for patient monitoring
Single and repeated measurements
Availability according to Countries or Regions:
The Suntech CT40 is a vital signs monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm and it is intended for bedside patient monitoring.
The technology used in the Suntech CT40, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
||Recent clinical validation; recent protocol
|BIHS (UK and IRL)
||Office and hospital use for adults (Preferred)
|AMA VDL (US)
||Published evidence plus 510(k) K160439
Polo Friz H, Punzi V, Petri F, Orlandi R, Maggiolini D, Polo Friz M, Primitz L, Vighi G. Simultaneous validation of the SunTech CT40 automated blood pressure measurement device by the 1993 British Hypertension Society protocol and the Association for the Advancement of Medical Instrumentation/International Organization for Standardization 81060-2: 2013 standard. Blood Press Monit. 2017 Oct;22(5):298-301. Epub: 2017 July 31. doi: 10.1097/MBP.0000000000000281. PMID: 28763332.
81060-2:2013 - Pass General population
BHS:1993 - Pass (A/A) General population