|BP||Medaval||★★★★ Certified Accuracy||Recent clinical validation; recent protocol with certified adherence|
|BP||BIHS (UK and IRL)||Previous recommendation, now archived||Published evidence|
|BP||ESH (Europe)||Self-measurement||Published evidence|
|BP||Japanese Society of Hypertension||Self-measurement (2016-2018)||Published evidence|
|BP||Stride BP||Self-measurement for adults (Preferred)||Published evidence|
Takahashi H, Yoshika M, Yokoi T. Validation of three automatic devices for the self-measurement of blood pressure according to the European Society of Hypertension International Protocol revision 2010: the Omron HEM-7130, HEM-7320F, and HEM-7500F. Blood Press Monit. 2015 Apr;20(2):92-7. doi: 10.1097/MBP.0000000000000096. PMID: 25462531.
ESH-IP:2010 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Omron HEM-7500F upper arm monitor, as validated according to the European Society of Hypertension International Protocol revision 2010. Medical Device Assessment. 2016 Aug 5;2016(1620) 5 p. Epub: 2019 Jan 31. Available from: www.medaval.ie.
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the Omron HEM-7500F, is certified by Medaval Ltd.