Brand: Omron Corporation, Kyoto Head Office, Shiokoji Horikawa, Shimogyo ku, Kyoto 600-8530, JAPAN.
Other: Omron Healthcare Manufacturing Vietnam Co. Ltd., 28 VSIP II, Street 2, Vietnam-Singapore Industrial Park II, Hoa Phu Ward, Thu Dau Mot City, Binh Duong Province 820000, VIET NAM.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The Omron HEM-7420 is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a 2-star Medaval rating. Blood pressure measurements are taken from the upper arm and it is intended for self-measurement and home use.
The technology used in the Omron HEM-7420, to measure blood pressure, has passed in a clinical validation study, in a general population, according to a recognised standard protocol, as published in a peer-reviewed publication.
||Recent clinical validation; recent protocol
||Self-measurement for adults (Preferred)
Topouchian J, Agnoletti D, Blacher J, Youssef A, Chahine MN, Ibanez I, Assemani N, Asmar R. Validation of four devices: Omron M6 Comfort, Omron HEM-7420, Withings BP-800, and Polygreen KP-7670 for home blood pressure measurement according to the European Society of Hypertension International Protocol. Vasc Health Risk Manag. 2014 Jan 16;10:33-44. doi: 10.2147/VHRM.S53968. PMID: 24476688. Available from: PMC3897354.
ESH-IP:2010 - Pass General population