Shanghai Honsun International Trade Co. Ltd., Room C. Floor 22, Zhiyuan Building, 768 Xietu Road, Shanghai 200023, CHINA.
Primary Client Use:
Intended for self-measurement and home use
Single measurements only
The Honsun Scian LD-735 is an automatic blood pressure monitor. Its blood pressure measurement technology has been proven to be accurate, with a certified 4-star Medaval rating. Blood pressure measurements are taken from the wrist and it is intended for self-measurement and home use.
The accuracy of the technology used in the Honsun Scian LD-735, to measure blood pressure, is certified by Medaval Ltd.
||★★★★ Certified Accuracy
||Recent clinical validation; recent protocol with certified adherence
|BIHS (UK and IRL)
||Self-measurement for adults
Kang YY, Chen Q, Li Y, Wang JG Validation of the SCIAN LD-735 wrist blood pressure monitor for home blood pressure monitoring according to the European Society of Hypertension International Protocol revision 2010. Blood Press Monit. 2016 Aug;21(4):255-8. Epub: 2016 April 18. doi: 10.1097/MBP.0000000000000192. PMID: 27093702.
ESH-IP:2010 - Pass General population
Medaval Certification Assessment
Medaval Ltd. Accreditation assessment of the blood pressure measurement technology used in the Honsun Scian LD-735 wrist monitor, as validated according to the European Society of Hypertension International Protocol revision 2010. Medical Device Assessment. 2016 Aug 5;2016(1630) 5 p. Epub: 2019 Jan 31. Available from: www.medaval.ie.
With proven validity of validation results according to an accepted protocol, the clinical accuracy of the blood pressure measurement technology, used in the Honsun Scian LD-735, is certified by Medaval Ltd.